Profesionální detektor štěnic-odposlechů, skrytých kamer + měřič frekvencí, IBQ101
- Výrobca: Meilin
- Kód produktu: 0040
- Dostupnosť: Skladom
- Bez DPH: 67,83 €
Digitálny multifunkčný merač frekvencií s mikroprocesorom a detektor
Môžete nájsť, zistiť prítomnosť a na akej frekvencii pracujú bezdrôtové zariadenia ako:
- skryté kamery
- diaľkové ovládače
- sieťové prvky
- otvárače dverí
- zabezpečenie domov, áut
- rušičky signálu a veľa ďalších.
- Frekvenčný rozsah 50 MHz až 2,6 GHz
- Veľký prehľadný 15-miestný a 6-riadkový LCD displej s menu a grafom a podsvietením
- Integrovaná nabíjačka batérií v merači + 3AAA nabíjacie batérie nejsou súčasťou
- Sila signálu Grafom zobrazujúci približovania sa k zdroju
- nízka spotreba
- H dokážete merať aj pri nabíjaní
- Rozmery 8,5 x 5,5 x 2 cm
- Hmotnosť 108g vrátane 3AAA batériou
- Nastavenie obnovovanie zobrazenia 0.1ms, 0.1s, 1s
- Odpor 50 Ohm
- indikátor batérie
- Auto dáta hold, podržanie hodnoty, zápis, zobrazenie
- Automatické vypínanie.
Nami uvedené zariadenie je NEPREDAJNÉ do súkromného sektora v SR. Len pre exportné účely.
Measuring range: 50MHz-2.6GHz
IBQ101 series handheld frequency meter is a series of ultra-small hand-held frequency measurement instruments, easy to operate and reliable performance.
Multi-function LCD screen display
Up to 9 digits display frequency
Analog digital dual signal strength indication
Measurement display remains
50 sets of measurement data saved
Human-computer interactive menu
With UV measuring antenna and USB power adapter
(1) Key functions
After the power is turned on, the current function of the three buttons of the panel is marked in the button function indicator column at the bottom of the display.
[LIGHT] Backlight button, press to turn the backlight on or off. After enabling the backlight, if there is no key operation for 5 consecutive minutes, the backlight automatically enters the power saving mode and automatically turns off the backlight.
[MENU] Enters the setup menu button.
[HOLD] Hold display key
[EXIT] Exit menu key
[SEL] selects the confirmation key and returns to the previous menu after selection
[NEXT] next selection key
[BACK] Return to previous menu key
(2). Turn on and off the power
In the off state, long press the F1 key for 2 seconds to enter the working state.
In working condition. Press and hold the F1 key for 2 seconds to display POWER OFF! Turn off the power.
(3) Backlight operation
In working condition. Press F1 to turn the backlight on. Press F1 again to turn off the backlight. After the frequency meter key has not operated for 5 minutes, the backlight display enters the power saving state and the backlight is automatically turned off.
(4) Keep the display operation
Press the F3 key, the frequency and signal strength of the display will remain on the display, the [HOLD] icon will be displayed in the title bar of the display. Press F3 again to close the display, and the [HOLD] icon in the title bar of the display will not be displayed. While keeping the display, the function of the F2 key changes from [MENU] to [SAVE]. Press the F2 key to save the frequency and signal strength in sequence. The title bar of the display [SAVE] flashes. After the saving is completed, the display title bar [SAVE] is not displayed, and the function of the F2 key returns to [MENU], keeping the display off.
(5) Set menu operation
Tap F2 to enter the setup menu [MENU].
There are three options in the settings menu, which are:
Measurement Range [RANGE] (102)
Meter Gate [GATE]
User settings [USER]
Press F1 to select the item to be set, press F2 to select to enter the next level menu, and press F3 to exit the setting menu.
(7) Meter gate setting [GATE]
Select [GATE] in the setup menu, press F2 to enter the [GATE] option, and press F3 to select the appropriate gate time. F1 exits.
(8) User settings [USER]
Hold keeps display on display
Save to save the measured frequency value
View View View Saved Frequency Values
Filter filter shows zero when no signal is tested
Autooff automatically turns off after turning on for 5 minutes
1) Keep the display setting [Hold]
Select [Hold] in the user settings
Press F3 to select [Auto] or [Manual]. When [Auto] is selected, press F2 to enter the hold level setup menu, and then tap F2 to increase F3 decrease. When [Manual] is selected for manual hold Press F1 or F2 to return to the previous setup menu.
2) Save Settings [Save]
Select [Save] in the user settings, select [Auto] for automatic save, and [Manual] for manual save.
To enable the auto-save function, first set the [Hold] function to [Auto]. When the measured signal strength exceeds the auto-hold level, the machine automatically saves the measured frequency and signal strength. The title bar of the display is saved when saving. The [SAVE] character flashes. After saving [SAVE] goes off.
Manually save. When in working state, press F3 to keep the display screen, F2 function is converted to manual save [SAVE], press F2 to manually save frequency and signal strength. The [SAVE] character flashes in the display title bar when saving. Save the [SAVE] off.
3) View saved data [View]
Select [View] in the user settings to display the frequency and signal strength, and the lower right corner of the data in the memory. Press F2 to see the next data. Press F3 to see the next data. Press F1 to return to [User].
4) Filter selection [Filter]
Select [Filter] in the user settings and select [ON] or [OFF]. The default setting is [ON]. The [Filter] function is turned off to maximize the sensitivity.
5) Auto off setting [AutoOff]
Select [AutoOff] in the user settings and select [ON] or [OFF] to enable or disable the auto power off function.
(9) Signal strength indicator
The signal strength indicator uses the analog bar and digital value to simultaneously display the measured signal strength. The signal strength indicator digital value can indicate the high-frequency signal with a resolution of 1dB. The port test has a maximum sensitivity of -40dBm and a test range of -40dBm to +15dBm. When using the antenna induction test method, the relative value of the frequency meter can only be used as a reference.
(1) Storage frequency meter
If you want to store the hand-held frequency meter for a long time, remove the battery inside the frequency meter before storing it. Release the battery cover cord to remove the battery.
(2) Charge the battery
When used for the first time, the built-in rechargeable Ni-MH rechargeable battery in the hand-held frequency meter may have insufficient power. Therefore, charge it before use. Insert the USB interface power adapter to charge the frequency meter. Note that this hand-held frequency meter does not have the function of connecting data with the PC, so it cannot be connected to the USB port of the PC. This USB port is only for charging and connecting to the COM port of the computer through the adapter.
When inserting the power adapter. The red LED in the upper right corner of the display is on, indicating that it is charging. The battery is fully charged and the green light is on.
Hand-held frequency meter can use the hand-held frequency meter's built-in battery to charge at the same time.
The battery indicator in the upper right corner of the handheld frequency meter screen indicates the usage of the built-in battery at that time. When the indicator flashes, it indicates that the battery is low. Please charge or replace the battery as soon as possible.
(3) Extend battery working time
When the auto-off function of the hand-held frequency meter is set to [ON], the frequency meter will automatically turn off to save power. If no button is pressed within 5 minutes, the frequency meter will automatically turn off. The frequency meter uses 3 AAANi-MH batteries for power supply. The frequency meter has a built-in Ni-MH battery charging circuit, which can effectively charge and manage the Ni-MH battery. In emergency situations, large-capacity alkaline batteries can also be used. However, when using alkaline batteries, do not attempt to charge alkaline batteries.
Enter the known standard signal, open the battery cover, remove the battery pack, use the charger to power the frequency meter, adjust the frequency reference, and make the displayed value equal to the standard signal.